26 June 1987
COMMERCIAL ITEM DESCRIPTION
The General Services Administration has authorized the use of this commercial item description.
This commercial item description covers a defibrillator analyzer to be used in testing of defibrillators and accompanying monitors when present.
The unit shall be portable unitized defibrillator/monitor analyzer.
All displays shall be digital.
The unit shall measure defibrillator output energy with a range maximum of at least 700 Joules, ± percent.
The defibrillator analyzer shall be compatible with Lown, Edmark and Trapezoid waveforms of up to 4kV amplitude.
The unit shall have a defibrillator pulse waveform memory.
The defibrillator analyzer shall have a real-time expanded time scale oscilloscope output, with calibrated 1000; 1 attenuation.
The defibrillator analyzer shall have an expanded time scale ECG recorder output with a time expansion factor of 200; 1.
The defibrillator analyzer shall measure and display actual cardioversion discharge delay time in milliseconds.
The unit shall measure defibrillator output energy via 50 ohm, ± 1 percent test load.
The unit shall have a defibrillator output energy test pulse.
The unit shall generate ECG performance waveforms to test monitor gain and damping adjustment, low frequency response, bandpass, band reject, high frequency response, linearity, and ECG rate meter/alarms.
DISTRIBUTION STATEMENT A. Approved for public release; distribution is unlimited.
For Parts Inquires submit RFQ to Parts Hangar, Inc.
© Copyright 2015 Integrated Publishing, Inc.
A Service Disabled Veteran Owned Small Business